When volume of media is as low as 100 ml in a mini SUB combined thermal control and mechanical support makes sense. The Heating-Support-Foot with the small CellVessel SUB may be mounted on most laboratory stirrer table or CerCell Magnetic-Stirrer-Table.
Electrical control of 500 ml SUB
IR radiation of HSF 2262-E
Heating-Support-Foot (HSF-E) for CellVessel integrate electrical cartridge heating element. Choose 48 VDC or 110 VAC or 230 VAC supply. Like p/n 2261-E-110 for use in US. (always including ground and to be used with ground!) All HSF for 500 ml SUB/SUF is OD 137 mm. HSF-E integrates a thermal protection which cut of at 55°C and connect again at 35°C. NOT for cultivation purposes – only for your personal protection! Drilled holes are 4,2 mm for SS well for OD 2,0 mm thermocouple and 6,2 mm for OD 6,0 mm thermocouple.
Liquid control of 500 ml SUB
HSF CellVessel 500 ml pn 2262-W
HSF pn 2262-W for 500 ml CellVessel
Heating-Support-Foot (HSF-W) for CellVessel 500 ml based on circulating liquid from an external supply. G1/4" connections and possible Rectus-21 couplings. Hollow aluminum construction. Drilled holes are one 4,2 mm one 6,2 mm for thermocouple (pn # 2262-W).
Mechanical support / thermal control of 500 ml SUB
HSF for size 500 mL pn 2262
two HSF for 500 mL pn 2262
The basic HSF (Heating-Support-Foot) (for your own electrical heating blanket) consisting of the aluminium ring shown here for the small CellVessel SUBs. Use the HSF for mechanical support, for thermal control with an electrical jacket of your own or our complete HSF. Drilled holes are: 4,2 mm and 6,2 mm for OD 6 mm thermocouple insert.
Check out numerous details under Support and Commerce